1) The device is DC high frequency portable dental x-ray unit, small, light and nearly no radiation.
2)There are manual buttons installed on the surface of shell, operate easily.
3)All of the components are installed in the central PC board, insulation vacuum and sealed stereotype protection make its brilliant features.
4)The unit mainly suitable for oral pre-treatment diagnosis of internal organization structure and root depth and so on, is indispensable in daily life clinic, especially for dental implant surgery.
5)The battery is durability, it can take about 500 images after full charged, and it can be charged and discharged for 1000 times in its life.
6)It can be connected with intra-oral digital x-ray imaging system.
Distance from skin to cone
Charger input voltage
4.1 The main body
1- Exposure switch
2- Beam cone
3- Control panel
4- Charging hole
5- Power switch
6- LCD screen
7- Indicate lights
8- Tooth position selection
9- Film/sensor selection
10- Adult/children selection
11- Exposure time adjusted
Warning About Using:
Make sure the person who using this device is legally.
If pregnant, ask a health professional before taking x-ray.
An excessive exposure of radiation leaking may cause human body reaction lightly.
Humidity conditions recommended for storage: 10-85%RH.
Humidity conditions recommended for use: 15-80%RH.
Optimal humidity range: 15-60%RH.
Temperature conditions recommended for storage: 10-50°C.
Temperature conditions recommended for use: 10-35°C.
Optimal temperature range: 10-25°C.
If any problem or error founded in the device, stop operating and then consult with the designated A/S center.
Do not disassemble or modify the device, since it include generates x-rays, if not use properly, it may damage users or patients.
Do not allow an unauthorized person to use or repair it.
The range of permissible voltage AC 100-240V, please use the charger within this range.
There is a potential danger for an electrical shock if you connect the power plug to consent or remove it.